Case Study: A BLE Communication Design of Glucose Monitor based on x73-PHD Standards and Continua Design Guidelines
International Journal of Wireless Communications and Mobile Computing
Volume 4, Issue 1, January 2016, Pages: 1-6
Received: Jan. 31, 2016;
Accepted: Feb. 12, 2016;
Published: Feb. 26, 2016
Views 5391 Downloads 100
Yuan-Fa Lee, Biomedical Technology and Device Research Laboratories, Industrial Technology Research Institute, Taiwan, R.O.C.
The ISO/IEEE 11073 (a.k.a. x73) specifications adopted by the Continua Health Alliance constitute the international personal healthcare device (PHD) communication standards. These standards allow medical devices to intercommunicate and exchange measurement data within a single system. The Bluetooth special interest group (SIG) introduced a Bluetooth Low Energy (LE) technology with reduced power consumption. The Bluetooth LE can be used in medical devices with low data transmission rates. The research aims to identify a practical solution for the Continua BLE glucose monitor. Based on the Bluetooth LE standards as well as the Continua design guidelines and x73-PHD standards, the proposed Continua BLE glucose monitor is developed. The Continua BLE glucose monitor takes glucose measurements from the user and transmits these data to the gateway via the Bluetooth LE interface, in compliance with the Bluetooth GATT-based data format and protocol standards. Our results show that glucose measurements can be successfully transmitted to an Android-based gateway and Continua-compliant gateway through the Bluetooth LE interface. The practical solution is also feasible for other medical devices, for instance, blood pressure monitor or thermometer.
Case Study: A BLE Communication Design of Glucose Monitor based on x73-PHD Standards and Continua Design Guidelines, International Journal of Wireless Communications and Mobile Computing.
Vol. 4, No. 1,
2016, pp. 1-6.
Copyright © 2016 Authors retain the copyright of this article.
This article is an open access article distributed under the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0/
) which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Y.-F., Lee and Y.-S. Huang, “Novel Personal Healthcare System,” 4th Intl. Symposium on Medical Info. and Communication Technology, 2010.
C.-Y. Park, J.-H. Lim, and S.-J. Park, “ISO/IEEE 11073 PHD Adapter Board for Standardization of Legacy Healthcare Device,” IEEE Intl. Conf. on Consumer Electronics, 2012.
Y.-F. Lee, “Personal Medical Monitoring System Based on x73-PHD Standards,” IEEE IT Professional, vol. PP, Issue 99, September 2012.
IEEE, Health Informatics–Personal Health Device Communication. Part 20601: Application Profile–Optimized Exchange Protocol, 2010.
Bluetooth SIG, Health Device Profile, 2008.
J. Kim and O. Song, “ISO/IEEE 11073 interoperability for person health device based on Zig Bee healthcare service,” IEEE Intl. Conf. on Consumer Electronics (ICCE), pp. 263-264, 2015.
Y.-J. Park, and H.-S. Cho, “Transmission of ECG Data with the Patch-Type ECG Sensor System using Bluetooth Low Energy,” Intl. Conf. on ICT Convergence (ICTC), pp. 289-294, 2013.
Z.-M. Lin, C.-H. Chang, N.-K. Chou, and Y.-H. Lin, “Bluetooth Low Energy (BLE) Based Blood Pressure Monitoring System,” Intl. Conf. on Intelligent Green Building and Smart Grid (IGBSG), pp. 1-4, 2014.
Continua Health Alliance, Continua Design Guidelines Version 4.0, 2013.
Bluetooth SIG, Bluetooth Specification Version 4.0, 2010.
Bluetooth SIG, Glucose Service, 2012.
Bluetooth SIG, Glucose Profile, 2012.
(2015, July.). Bluetooth low energy software stack and tools. [Online]. Available: http://www.ti.com/product/CC2540/toolssoftware
(2015, July.). Continua Enabling Software Library (CESL). [Online]. Available: https://cw.continuaalliance.org/wg/members/home/cesl-download.
(2015, July.). Continua Test Tool. [Online]. Available: https://cw.continuaalliance.org/wg/members/home/test-tool-download.