Assessment of Radiation Exposure of Head and Neck during Panoramic Radiography in Different Age Groups
Science Journal of Clinical Medicine
Volume 3, Issue 1, January 2014, Pages: 10-14
Received: Jan. 3, 2014;
Published: Feb. 20, 2014
Views 3015 Downloads 211
Ehsan Moudi, Department of Oral and Maxillofacial Radiology, Babol University of Medical Sciences, Babol, Iran
Sina Haghanifar, Department of Oral and Maxillofacial Radiology, Babol University of Medical Sciences, Babol, Iran
Hoora Hadian, Post Graduate Student of Oral and Maxillofacial Radiology, Shahid Beheshti University of Medical Sciences, Tehran, Iran
Ali Shabestani Monfared, Department of Biochemistry and Biophysics, Babol University of Medical Science
Gholamreza Deilam, Private Dentist, Babol, Iran
Nika Bahemmat, Post Graduate Student of Oral and Maxillofacial Radiology, Shahid Beheshti University of Medical Sciences, Tehran, Iran
Background and aim: Panoramic imaging is one of the most commonly used imaging techniques in dentistry. Being able to accurately assess the radiation dose patients receive during procedures is a crucial step in the management of dose. The main objective of this study was to evaluate the head and neck skin absorbed dose during panoramic radiography in different age groups with panoramic machine of Oral and Maxillofacial Radiology section of Babol school of dentistry. Materials and methods: 273 thermoluminescent dosemeters (TLDs) (100 LiF: Mg, Ti , harshaw, USA) were used.90 samples were selected from the patients who referred to the Oral and Maxillofacial Radiology section of Babol school of dentistry for panoramic radiography. .Samples divided in 3 age_ group: 4_ 10 years, 10_40 years and above 40 years . TLDs were calibrated in dosimetry laboratory, National Radiation Protection Department. Thermoluminescent signal was read out with a Harshaw 4500 (Harshaw, Bicron USA) reader. Mean and standard deviation was determined by SPSS10 software and ANOVA statistic analysis. Results: Mean ± SD of skin absorbed dose of head and neck for 90 patients was 0.47± 0.09mGy. Conclusions: Since Diagnostic Reference level (DRL) of panoramic imaging is unknown in Iran, there is no possibility to compare the current results with DRL. However, This study conclude that decrease of radiation dose seems to be achievable with lower exposure condition with the panoramic unit.
Ali Shabestani Monfared,
Assessment of Radiation Exposure of Head and Neck during Panoramic Radiography in Different Age Groups, Science Journal of Clinical Medicine.
Vol. 3, No. 1,
2014, pp. 10-14.
Ribeiro DA. Cytogenetic biomonitoring in oral mucosa cells following dental X-ray. Dentomaxillofac Radiol. 2012 Mar;41(3):181-4.
Batista WO, Navarro MV, Maia AF.Effective doses in panoramic images from conventional and CBCT equipment.Radiat Prot Dosimetry. 2012;151(1):67-75. Epub 2011 Dec 14.
Williams JR, Montgomery A.Measurement of dose in panoramic dental radiology.Br J Radiol. 2000 Sep;73(873):1002-6.
Zhang Y, Li X, Segars WP, Samei E. Organ doses, effective doses, and risk indices in adult CT: comparison of four types of reference phantoms across different examination protocols. Med Phys. 2012 Jun;39(6):3404-23.
Ludlow JB. Comment on "Effective dose range for dental cone beam computed tomography scanners".Eur J Radiol. 2012 Dec;81(12):4219-20.
S.M.J. Mortazavi , M. Ghiassi-Nejad , M. Bakhshi , M. Jafari-Zadeh , A. Kavousi , J. Ahmadi , et al. Entrance Surface Dose Measurement on the thyroid gland in orthopantomography: The need for optimization. Iranian Journal of Radiation Research. 2004; 2 (1) :21-26.
[No authors listed]Radiological protection and safety in medicine. A report of the International Commission on Radiological Protection.Ann ICRP. 1996;26(2):1-47.
[No authors listed]Diagnostic reference levels in medical imaging: review and additional advice. Ann ICRP. 2001;31(4):33-52
Asadinezhad M, Bahreyni Toossi MT. Doses to patients in some routine diagnostic X-ray examinations in Iran: proposed the first Iranian diagnostic reference levels. Radiat Prot Dosimetry. 2008;132(4):409-14. Doi: 10.1093/rpd/ncn308. Epub 2008 Dec 23.
Toosi MT, Asadinezhad M. Radiat Prot Dosimetry. Local diagnostic reference levels for some common diagnostic X-ray examinations in Tehran county of Iran.2007;124(2):137-44. Epub 2007 May 10.
Faghihi R, Mehdizadeh S, Sina S, Alizadeh FN, Zeinali B, Kamyab GR, Aghevlian S, Khorramdel H, Namazi I, Heirani M, Moshkriz M, MahaniH, SharifzadehM. Radiation dose to neonates undergoing X-ray imaging in special care baby units in Iran.Radiat Prot Dosimetry. 2012 Jun;150(1):55-9. Doi: 10.1093/rpd/ncr373. Epub 2011 Nov 28.
White SC, Pharoah MJ. Oral radiology : principles and interpretation. 6th ed. New Delhi : Mosby, 2009 :31.87.
Gijbels F, Jacobs R, Bogaerts R, Debaveye D, Verlinden S, Sanderink G. Dosimetry of digital panoramic imaging. Part I: Patient exposure. Dentomaxillofac Radiol. 2005 May;34(3):145-9.
Doyle P, Martin CJ, Robertson J.Techniques for measurement of dose width product in panoramic dental radiography.Br J Radiol. 2006 Feb;79(938):142-7.
Garcia Silva MA, Wolf U, Heinicke F, Gründler K, Visser H, Hirsch E. Effective dosages for recording Veraviewepocs dental panoramic images: analog film, digital, and panoramic scout for CBCT. Oral Surg Oral Med Oral Pathol Oral Radiol Endod. 2008 Oct;106(4):571-7. Epub 2008 Jul 7.
Lorenzoni DC, Bolognese AM, Garib DG, Guedes FR, Sant'anna EF.Cone-beam computed tomography and radiographs in dentistry: aspects related to radiation dose. Int J Dent. 2012;2012:813768. Epub 2012 Apr 4.
Gavala S, Donta C, Tsiklakis K, Boziari A, Kamenopoulou V, Stamatakis HC.Radiation dose reduction in direct digital panoramic radiography. Eur J Radiol. 2009 Jul;71(1):42-8. doi: 10.1016/j.ejrad.2008.03.018. Epub 2008 May 2.
[Article in Japanese]Matsuo A, Okano T, Gotoh K, Yokoi M, Hirukawa A, Okumura S, Koyama S. [Absorbed dose and the effective dose of panoramic temporo mandibular joint radiography]. Nihon Hoshasen Gijutsu Gakkai Zasshi. 2011;67(10):1275-83.
Grünheid T, Kolbeck Schieck JR, Pliska BT, Ahmad M, Larson BE. Dosimetry of a cone-beam computed tomography machine compared with a digital x-ray machine in orthodontic imaging. Am J Orthod Dentofacial Orthop. 2012 Apr;141(4):436-43.