American Journal of Physics and Applications
Volume 2, Issue 5, September 2014, Pages: 104-107
Received: Sep. 19, 2014;
Accepted: Sep. 29, 2014;
Published: Sep. 30, 2014
Views 3409 Downloads 265
Seyed Ali Rahimi, Health Sciences Research Center , Mazandaran University of Medical Sciences, Kilometer 18 Khazarabad Road, P.O.BOX:48175-1553, Sari, Iran
There are analytic methods for designing protective barriers however, they lack sufficient efficiency and considering the NCRP (National Council on Radiation Protection and measurements) reports, designing mechanical protective barrier in order to protect the initial x-ray radiation and absorption of the ray quality of such radiation is different. In this study, computer software was designed to calculate the needed barrier with high accuracy. For proper determination of thickness of the protective barrier, relevant information about curves of radiation weakness, dose limit and other items should be entered. This program was done in windows and designed in such a way that the operator works easily, flexibility of the program is acceptable and its accuracy and sensitivity is high. Meanwhile sometimes shielding is more than what required which lacks technical standards and cost effectiveness. When the application index is contrasting zero, thickness of NCRP49 calculation is about 20% less than the calculated rate done by the method of this study. The multi radiation sources in a single room are considered and non guaranteed radiation of NCRP hypothesis is removed. Difference between the theoretical and calculated rates of this method is X2 =10-5 which indicates accuracy and high efficiency of this software.
Seyed Ali Rahimi,
Construction Shielding Design for Medical X-Ray Imaging Equipment, American Journal of Physics and Applications.
Vol. 2, No. 5,
2014, pp. 104-107.
Benjamin R. archer, T.R. fewell ,B. J. Conway and P.w. quinn "Attenuation properties of diagnostic x-ray shielding materials" Health physics Vol. 21, No.9. September, pp, 1499-1505, 1994.
NCRP Report 49, " structural Shielding Design and evaluation for medical use of x-ray rays and gamma rays of energies up to 10 mev" , national council on radiation protection and measurements, 1976.
Edward L. Nickoloff and Eileen M. Donnelly"Use of gypsum drywall as shielding material for mammography" Health physics Vol.54, No.6,(June) , pp. 465-468,1989.
C. Maccia, M Benedittini and C. lefaure, f. fagnani "Doses to patients from diagnostic radiology in france" Health physics Vol.54, No.4, (April), pp. 397-340, 1988
Stewart C.bushong and Sharon A .glaze "Radiographic workload and use factors for orthopedic facllties" Health physics Vol.44, NO.1, (January). Pp.53-59, 1983
G.Panagiotakis, J. kandarakis, "Lecture notes on diagnostic radiology for the msc in European medical physics",1995.
International commission radiological units ,"Quantities and units in radiation protection dosimetry ", ICRU Report 51.
International commission on radiological protection, "1990 recommendations of the international commission on radiological protection," ICRP publication 60.
M.petrantonaki,C,kappas, E.P. Efstathopiulos, Y, theodorakos, GlPanagiotakis, "Calculating shielding requirements in diagnostic x-ray departments", August 1998
ICRP Publication 26 (1977), "recommendations of the international commission on radiological protection".
Seyed Ali Rahimi,EVALUATION of RADIOLOGY PERSONNEL PRACTICE OF Mazandaran UNIVERSITY OF MEDICAL SCIENCES HOSPITALS. QANTRM(2006)-Organized by the ,IAEA- Vienna- Austria.
Seyed Ali Rahimi, Study on the performance of recommended standards in the diagnostic radiology units of the hospitals affiliated to the Mazandaran University of Medical Scienc,Journal of X-Ray Science and Technology - Amsterdam Ios press- 15(2007)57-63- Netherlands.
Methé BM. Shielding design for a PET imaging suite: a case study. Health Phys. 2003 May; 84(5 Suppl):S83-8.
Casanova AO, López N, Gelen A, Guevara MV, Díaz O, Cimino L, D'Alessandro K, Melo JC. Shielding analysis of the Microtron MT-25 bunker using the MCNP-4C code and NCRP Report 51. Radiat Prot Dosimetry. 2004;109(3):189-95.
Fog LS, Cormack J. Mathematical modeling of the radiation dose received from photons passing over and through shielding walls in a PET/CT suite. Health Phys. 2010 Dec;99(6):769-79. doi: 10.1097/HP.0b013e3181e47a39.
Fujita Y, Myojoyama A, Saitoh H. BREMSSTRAHLUNG AND PHOTONEUTRON PRODUCTION IN A STEEL SHIELD FOR 15-22-MEV CLINICAL ELECTRON BEAMS. Radiat Prot Dosimetry. 2014 May 12.
Mori H, Koshida K, Ishigamori O, Matsubara K. A novel removable shield attached to C-arm units against scattered X-rays from a patient's side. Eur Radiol. 2014 Aug;24(8):1794-9.
Colletti PM, Micheli OA, Lee KH. To shield or not to shield: application of bismuth breast shields. AJR Am J Roentgenol. 2013 Mar;200(3):503-7.
Elschot M, de Wit TC, de Jong HW. The influence of self-absorption on PET and PET/CT shielding requirements. Med Phys. 2010 Jun;37(6):2999-3007.